My Zeiss Jena LgOG1
(Myself new to microscopy)
The microscope had an earlier life at an educational institution in Oranienburg, Germany as number 7. Then somehow ended up at a german fleamarket, brought out of Germany and after some time placed on the online market and subsequently brought to my doorstep.
Have most likely had a quiet life, not many scratches. Movable above the square mechanical stage but mostly static below. Solidified oil/grease.
Seems like someone has used to much force on one of the delicate screws controlling the square mechanical stage. Time will show.
No sooner than just becoming a member MicroBob were there providing valuable help and documentation for solving the 'solidified' problem. Much appreciated.
File comment: As received
Zeiss Jena LgOG1.jpg [ 32.48 KiB | Viewed 2180 times ]
It came in a cabinet:
File comment: LgOG1 Cabinet
LgOG1 Cabinet.jpg [ 40.1 KiB | Viewed 2180 times ]
Wonder what equipment the two installed 'holders' left and upper right were ment for?
File comment: LgOG1 Cabinet loose lock part
LgOG1 Cabinet loose lock part..jpg [ 15.87 KiB | Viewed 2180 times ]
Wonder what the missing key looked like? I should try to get/make a new one!
File comment: LgOG1 Huygenian eyepieces
LgOG1 Huygenian eyepieces.jpg [ 54.72 KiB | Viewed 2180 times ]
File comment: LgOG1 Tools
LgOG1 Tools.jpg [ 36.17 KiB | Viewed 2180 times ]
Where to use the four small Tools at the right?
Where and how to use the "stützplatte"?
I know how to use the distance part for the light system but are in need of the light source itself, (probably from the german ebay).